§ 221.28. Technique indicators.
(a) The technique factors for radiographic systems shall be indicated before exposure except for units utilizing automatic exposure controls, in which case the maximum mAs shall be indicated.
(b) The requirement of subsection (a) may be met by permanent markings on equipment having fixed technique factors. Indication of technique factors shall be visible from the operators position except in the case of spot films made by a fluoroscopist.
The provisions of this § 221.28 issued under sections 301 and 302 of the Radiation Protection Act (35 P. S. § § 7110.301 and 7110.302); and section 1920-A of The Administrative Code of 1929 (71 P. S. § 510-20).
The provisions of this § 221.27 adopted December 18, 1987, effective December 19, 1987, 17 Pa.B. 5235; amended October 2, 1998, effective October 3, 1998, 28 Pa.B. 4894. Immediately preceding text appears at serial page (123686).
This section cited in 25 Pa. Code § 221.35a (relating to fluoroscopic X-ray systems).
No part of the information on this site may be reproduced for profit or sold for profit.
This material has been drawn directly from the official Pennsylvania Code full text database. Due to the limitations of HTML or differences in display capabilities of different browsers, this version may differ slightly from the official printed version.