§ 221.33a. Radiation from capacitor energy storage equipment in standby status.
Radiation emitted from an X-ray tube when the exposure switch or timer in not activated may not exceed a rate of 2 milliroentgens (0.516 µC/kg) per hour at 5 centimeters from an accessible surface of a fully charged diagnostic source assembly, with the beam-limiting device fully open.
The provisions of this § 221.33a issued under sections 301 and 302 of the Radiation Protection Act (35 P. S. § § 7110.301 and 7110.302); and section 1920-A of The Administrative Code of 1929 (71 P. S. § 510-20).
The provisions of this § 221.33a adopted October 2, 1998, effective October 3, 1998, 28 Pa.B. 4894; amended November 16, 2001, effective November 17, 2001, 31 Pa.B. 6282. Immediately preceding text appears at serial pages (249295) to (249296).
This section cited in 25 Pa. Code § 221.35a (relating to fluoroscopic X-ray systems).
No part of the information on this site may be reproduced for profit or sold for profit.
This material has been drawn directly from the official Pennsylvania Code full text database. Due to the limitations of HTML or differences in display capabilities of different browsers, this version may differ slightly from the official printed version.