§ 221.41a. Fluoroscopic timer.
A cumulative timing device activated by the fluoroscope switch shall be provided. It shall indicate the passage of a predetermined period of irradiation either by an audible signal or by temporary or permanent interruption of the irradiation when the increment of exposure time exceeds a predetermined limit not exceeding 5 minutes.
The provisions of this § 221.41a issued under sections 301 and 302 of the Radiation Protection Act (35 P. S. § § 7110.301 and 7110.302); and section 1920-A of The Administrative Code of 1929 (71 P. S. § 510-20).
The provisions of this § 221.41a adopted October 2, 1998, effective October 3, 1998, 28 Pa.B. 4894.
This section cited in 25 Pa. Code § 221.43a (relating to mobile fluoroscopes); and 25 Pa. Code § 221.61 (relating to radiation therapy simulation systems).
No part of the information on this site may be reproduced for profit or sold for profit.
This material has been drawn directly from the official Pennsylvania Code full text database. Due to the limitations of HTML or differences in display capabilities of different browsers, this version may differ slightly from the official printed version.